[time-nuts] Have done some more cutting on the Cs beam tube
paulswedb at gmail.com
Sun Feb 19 18:38:51 EST 2017
Really enjoy the pictures and the high resolution captures. Learning a lot.
I guessed the c field was the bottom coil and the small coil in the bean
was the LF coil. But it clearly explains the low z of the LF coil. Boy
thats about a direct short. :-)
With respect to the C Field Toms pix help make more sense of the wiring.
On Sun, Feb 19, 2017 at 7:38 AM, Richard (Rick) Karlquist <
richard at karlquist.com> wrote:
> On 2/18/2017 7:43 PM, Tom Van Baak wrote:
>> Unless yours is a high-perf 5061 tube I wonder if the degaussing coil
>> even exists. Did you pull it from an option /004 frame? Check if there's a
>> separate pair of degaussing wires coming from the tube and heading to the
>> rear panel of the instrument. That would make a total of 6 coil wires.
>> Corby will know all of this for sure.
> I remember that the degausser product that drove the
> degaussing coil was a very low priority because Len
> Cutler et al insisted that it was unnecessary to
> degauss the CBT, at least in the 5071A. It finally
> got produced just to satisfy a few customers who
> believed they wanted it.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/m
> and follow the instructions there.
More information about the time-nuts