[time-nuts] Nortel NTGS and LH 5
bpl521 at outlook.com
Fri Feb 24 21:37:14 EST 2017
Thanks Mark, /ro=0 did the trick. It did leave a "r" at the end of the corrected date, but after a restart of LH it is now just displaying the date. Thanks again.
From: time-nuts <time-nuts-bounces at febo.com> on behalf of Mark Sims <holrum at hotmail.com>
Sent: February 23, 2017 8:48 AM
To: time-nuts at febo.com
Subject: [time-nuts] Nortel NTGS and LH 5
Lady Heather has detected a GPS week rollover event... 10 seconds of consecutive years less than 2016. When this happens Heather adds 1024 weeks to the receiver date until it is past 2016. This can happen when Heather is connected to a receiver that is starting up and sending invalid dates or a receiver with rollover problems. It can also happen with a receiver that reset or glitched while Heather was running. You can force the rollover offset with the /ro=seconds_to_add command /ro=0 will effectively disable rollover compensation.
> Perhaps something just went wonky in my setup but just noticed that the date that LH 5 (TSIP) is displaying with my Nortel is very wrong. Any idea how to correct. Could have happened after I did a cold reset a couple weeks ago.
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
time-nuts Info Page - American Febo Enterprises<https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts>
time-nuts is a low volume, high SNR list for the discussion of precise time and frequency measurement and related topics. To see the collection of prior postings to ...
and follow the instructions there.
More information about the time-nuts