[time-nuts] Trimble Thunderbolt no longer determines the correct date
cubecentral at gmail.com
Sat Jul 29 20:30:29 EDT 2017
It happened here too. It is reporting having the 3.0 firmware. Right at the time Tom said. Fortunately I am simply using this unit for PPS and nothing else.
From: time-nuts [mailto:time-nuts-bounces at febo.com] On Behalf Of Tom Van Baak
Sent: Saturday, July 29, 2017 18:16
To: Discussion of precise time and frequency measurement <time-nuts at febo.com>
Subject: Re: [time-nuts] Trimble Thunderbolt no longer determines the correct date
Caught it. Some Trimble Thunderbolt TBoltmon.exe screen shots attached:
GPS WN 1959 TOW 604799 (July 29, 2017 23:59:41) advanced to GPS WN 936 TOW 0 (December 13, 1997) instead of GPS WN 1960 TOW 0 (July 29, 2017 23:59:42).
1960 - 936 is 1024 weeks, as advertised for this version of the TBolt GPSDO. Note this happened at 23:59:42 UTC as expected (that's GPS midnight - 18 UTC leap seconds). I did not expect the reported 2.75 us 1PPS phase change and will look into that.
More information about the time-nuts