[time-nuts] HP 5065A repair...
paulswedb at gmail.com
Sun Jul 30 19:24:14 EDT 2017
Ulf very good and great to hear. I did look at the clock motor driver and I
believe you can use a ttl flip flop to drive the two transistors just fine.
I think the ttl will run on the 4.2 V. But the transistor drive
requirements look reasonable.
So if you want the click click you can have it.
On Sun, Jul 30, 2017 at 6:28 PM, Ulf Kylenfall via time-nuts <
time-nuts at febo.com> wrote:
> After having salvaged an old 5065A that was
> decomissioned in 1987 due to Rubidium cavity
> heater short circuit, I have now, several
> weeks later eventually managed to what I think
> repair it.
> Using a GPS-Diciplined HP105A
> as oscilloscope trigger, the 5MHz O/P is
> absolutely still. "Continous Operation"is ON, and the values indicated on
> thepanel meter are almost matching thosewritten on the operating chart back
> in 1987.
> The next step is to take the 5065 to my work
> where we have two H-masers. Then it will be possible
> to see if it meets the old specification.
> A BIG Thank You
> to Corby Dawson!for valuable help and also to people providing
> the documents found on KO4BB site ("First Aid...")
> Ulf KylenfallSM6GXV
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/
> and follow the instructions there.
More information about the time-nuts