[time-nuts] Thunderbolt Rollover Testing
holrum at hotmail.com
Sun Mar 5 19:38:35 EST 2017
Does it have a saved/surveyed position? With a saved position you can reasonable time performance with 1 sat. Without a saved position all bets are off, there is no way for the receiver to determine the receiver/satellite clock difference.
Trimble reports that the device is in "over-determined clock" mode if that is how it is configured. Even if it is in "over-determined clock" mode the receiver may not have enough data to do anything effective. Trimble's overdetermined clock mode is basically a "position hold" mode.
>I just re-checked this with a Resolution-T, and it does go into OD mode with all but 1
SV masked from cold restart using a real SV.
More information about the time-nuts