[time-nuts] Fwd: Dusty 53131A
magnus at rubidium.dyndns.org
Mon Mar 6 05:12:02 EST 2017
To state the obvious, I think the same goes for 53132A and indeed
On 03/06/2017 08:19 AM, Bob Camp wrote:
> 53131’s are dust magnets. It is not at all uncommon for them to fill up with dust,
> overheat, and die. Popping them open every 10 years or so to clean them out is
> well worth the effort.
>> On Mar 5, 2017, at 8:54 PM, Scott Stobbe <scott.j.stobbe at gmail.com> wrote:
>> Recently the fan in my 53131 has been making a pretty awful sound, so its
>> time to replace it. When I disassembled the counter I found the trim cap
>> had turned into a fur ball. Not that it really matters since it runs on an
>> external reference, but its interesting to see if the dust has any effect.
>> See https://www.flickr.com/photos/147407087@N06/32427230764/. So with the
>> unit disassembled I logged the effect of removing the dust by logging a
>> GPSDO during the cleaning process.
>> Must of the dust was around the shroud of the trim cap. Near the mounting
>> terminals there was fairly minimal amounts of dust. At least in this
>> instance, if the dust had any impact on frequency, its orders of magnitude
>> below the thermal instability of the AT crystal.
>> The crystal appears to be in a to-39 3-lead package. It is also
>> freestanding directly in front of the exhausting cooling fan...
>> time-nuts mailing list -- time-nuts at febo.com
>> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
>> and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts