[time-nuts] Fwd: HP10811 Oscillator Thermal Fuse
David G. McGaw
david.g.mcgaw at dartmouth.edu
Wed May 10 23:20:42 EDT 2017
I too have had a fuse open up due to a failed thermistor in a HP10811.
On 5/10/17 7:43 PM, Dan Rae wrote:
> As I reported the last time this subject came up, or maybe the time
> before, the only time I had an open thermal fuse was in a 10811 that
> had an open thermistor. I was able to replace that and get the OCXO
> working, but if the fuse had been replaced with a wire link I'm sure
> the entire oven would have been toast.
> There have been two types of fuse fitted, all I those have are the
> type that can be fitted with the plug in axial panasonic type, less
> than a dollar from Digikey. The older ovens had a different type but
> I imagine it's similarly easy to replace.
> All my 10811s have the higher temperature Panasonic fuse fitted and I
> sleep well at night...
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts