[time-nuts] Some more info on the 5065A optical unit
kb8tq at n1k.org
Mon Nov 27 12:47:03 EST 2017
If the coil form really is plain old cardboard. I wonder what the impact
of humidity is? With the heat rise and all the packing, it’s going to be
a pretty indirect effect. It might be an issue on a unit that had been stored
powered off for a number of years. Straight cardboard *is* an issue on
RF coils in humidity.
> On Nov 27, 2017, at 12:34 PM, cdelect at juno.com wrote:
> Made some measurements on a 5065A optical unit.
> -There are 3 layers in the magnetic shield.
> -There is about 1/8" of insulation between the
> outer and middle, and between the middle and inner shields.
> -The C-field winding is wound on a cardboard form that
> fits tightly into the inner shield.
> -There is almost 1/2" of foamed insulation between the C-field
> winding and the cell and oven forms and windings.
> Per Poul-Hennings data a 3.25 deg C change of the outer shield
> will result in a 1 deg C change in the C-field coil.
> This equates to a 3.4X10-14 frequency shift when not using
> active current drive.
> If the outer shield is actively held to plus/minus 1 deg C then
> the shift will be held within plus/minus 1.05X10-14 as long as
> the ambient temperature stays within the regulating range of the
> PC liquid cooler.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts