[time-nuts] Rakon HSO-14
John Ackermann N8UR
jra at febo.com
Sun Feb 4 14:54:05 EST 2018
Here's a link to data on an 8607-008. You may recognize the bottom plot
from a recent posting. :-) But the ADEV and PN data at the top of the
page is from the factory test data. The ADEV doesn't explicitly say so,
but I strongly believe it's based on comparison with a "gold standard" 8607.
On 02/04/2018 02:38 PM, Bob kb8tq wrote:
> Obviously you need two more 8607’s ….. :)
> I suspect you are correct and the OCXO is doing better than the close in data suggests.
>> On Feb 4, 2018, at 8:33 AM, Tom Van Baak <tvb at LeapSecond.com> wrote:
>> FYI: here's an old plot where I evaluated an Oscilloquartz 8607 BVA against a H-maser. It gets down to 8e-14 but is likely a bit better. On this plot, I suspect the short-term numbers were not the BVA oscillator or the TSC 5110A analyzer, but the H-maser.
>> time-nuts mailing list -- time-nuts at febo.com
>> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
>> and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts