[time-nuts] Does a frequency counter locked to GPS need to "warm up"?
azelio.boriani at gmail.com
Thu Feb 8 03:34:36 EST 2018
No warmup needed if the counter switches to the external reference. If
the counter uses the external reference to lock the internal one then
it is better to wait until the internal one is stable, maybe the
counter has a standby mode where it appears powered off but the
internal reference is active (usually an ovenized reference). For the
best performance (stability of trigger levels, input amplifiers and so
on) it is better to warmup the whole counter always.
On Thu, Feb 8, 2018 at 8:33 AM, Chris Wilson <chris at chriswilson.tv> wrote:
> 08/02/2018 07:31
> Does a frequency counter connected to a permanently running (Trimble Thunderbolt) GPS
> disciplined frequency standard need to warm up after switch on before readings settle?
> Just curious, thanks.
> Best Regards,
> Chris Wilson.
> mailto: chris at chriswilson.tv
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts