[time-nuts] RE CSAC Project(was CSAC purchase)
ronaldheld at gmail.com
Tue Jan 23 19:40:16 EST 2018
Good to keep in mind before I decide what to do.
The original RbXO was not a whole lot bigger than the Rb. The CSAC
is a *lot* smaller than the Rb’s of that era. The TCXO’s and OCXO’s today
are a lot smaller as well. I’d bet you could do it in < 2X the volume of the
CSAC with a pretty good OCXO and < 1.2X the volume with a precision TCXO.
On 1/23/18, Ronald Held <ronaldheld at gmail.com> wrote:
> What is you idea of portable in terms of size and mass for RbXO?
> I suppose I am try to do "better' and my TCXO watches which at best
> run around a second/year.
> OCXO and TCXO are both available smaller than the CSAC (particularly
> tcxo). I'm using a vectron EX-421 OCXO and it's about 1cm on a side,
> the OX205 is about 1" square and maybe 0.60" tall.
> TCXOs are available in "cellphone" form factors (e.g. tiny SMT packages)
> Long term, maybe a year, sounds like a reasonable goal. Maybe I am
> just chasing the next zero, if I have the metaphor correct?
> I’m guessing there was a question to me that somehow got lost in the world
> ones and zeros ….
> My comment was in terms of temperature stability. The CSAC has a temp
> specification of +/-4x10^-10 over -10 to +70C. There are TCXO’s that
> will get below
> 5x10^-9 over that range and use far less power. There are OCXO’s that will
> to better temperature stability numbers over that range. Neither one
> will do the long
> term aging that a Rb will.
More information about the time-nuts