[time-nuts] More on SiLabs 5340
attila at kinali.ch
Tue Jan 30 15:16:41 EST 2018
On Mon, 29 Jan 2018 15:38:33 -0500
John Ackermann N8UR <jra at febo.com> wrote:
> Today I tried replacing the crystal with a 10 MHz input, and found that
> the evaluation board requires surgery to support that. What I was able
> to do, though, was feed 10 MHz into the "Input 0" port which apparently
> disciplines the free-running crystal. I used a 10811A fed into a
> T2-Mini with divider removed, so all it does is provide the Wenzel
> squaring circuit. Results attached, added to the plots from yesterday.
> Also attached is an ADEV plot proving that the 10 MHz is in control.
There is a small detail that puzzles me: the ADEV for the internal
reference 10MHz and 30MHz bends upwards at tau greater than 0.1s.
Shouldn't that be visible by a change in slope in the phase noise
plot at ~10Hz as well?
It is upon moral qualities that a society is ultimately founded. All
the prosperity and technological sophistication in the world is of no
use without that foundation.
-- Miss Matheson, The Diamond Age, Neil Stephenson
More information about the time-nuts