[time-nuts] Thunderbolt code phase measurement
michaeljwouters at gmail.com
Wed May 16 03:19:45 EDT 2018
By "hosing" do you mean that you lose messages for the next second? That
was a problem with the Resolution T too. I wanted to get ephemerides from
the receiver so I ended up minimising lost messages by tracking satellites
as they popped into view, and then requesting an ephemeris after a suitable
wait. This problem went away in the SMT 360.
Regard resolving the ms ambiguity, you could look at the code in mktimetx,
which does just what Magnus D describes.
The code lives at
On Wed, 16 May 2018 at 4:02 pm, Mark Sims <holrum at hotmail.com> wrote:
> Many thanks Peter for confirming what I suspected. The problem with the
> Trimble receivers is that requesting the satellite C/A code data can hose
> up a lot of them. So, I'm stuck with calculating the integer number of
> milliseconds. How to do that? I do know my position to a few feet.
> I have Lady Heather generating RINEX files for the Ublox timing receivers,
> the NVS-08, the Furuno GT-87, and the Ashtech Z12 (with both L1/L2 data).
> It would be nice to be able to support the Trimble receivers. With L1
> only data I am getting results in the < 200 mm range. The Z12 with L1/L2
> data gets me to around 40 mm.
> > If you know your position to within 150 kilometers (0.5 ms), you can
> dispense with the pseudorange-assembly arithmetic and just use the code
> phase directly, after adding in the appropriate integer number of
> milliseconds, only one of which will put you within your known
> 300-kilometer-diameter (1 ms) sphere.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts