[time-nuts] HP 5372A TIA Question
ed_palmer at sasktel.net
Thu May 31 12:15:06 EDT 2018
On 5/31/2018 10:00 AM,
"Poul-Henning Kamp"<phk at phk.freebsd.dk> wrote:
> PS: I do not recall it being mentioned, but the 1 MOhm input pods are
> horribly unstable compared to the 50 Ohm input pods, which are
> basically just a stripline and a BNC connector.
One thing to note about all the 5372A pods: They should all be checked
for bad solder joints between the rear connector and the circuit
boards. In my collection of pods of all three types, almost all had one
or more broken solder joints.
More information about the time-nuts