[volt-nuts] History of HP 3456A
richiem at hughes.net
Tue Nov 23 19:25:31 UTC 2010
Thanks, Marv, for putting this together. RE personal experiences, I've owned two 3456As and never had a probelm except one of them had front panel tactile swittches that kept losing their springs down into the bottom of the chassis. I've also owned 2 3457As and in my opinion, the 3456 was a FAR better meter in regards to stability of calibration and ease of use, not to mention clarity of readouts.
On Nov 23, 2010, at 4:00 AM, volt-nuts-request at febo.com wrote:
> Message: 1
> Date: Tue, 23 Nov 2010 06:39:22 -0500
> From: "Marvin E. Gozum" <marvin.gozum at jefferson.edu>
> Subject: [volt-nuts] A history of the HP 3456A
> To: volt-nuts at febo.com
> Message-ID: <22.214.171.124.2.20101123063905.04874970 at comcast.net>
> Content-Type: text/plain; charset="us-ascii"; format=flowed
> I'm collecting a history of this DVM.
> Does anyone have any other resources, know of specifics, can check
> the accuracy of what I've already listed, and have personal anecdotes
> of their experience with this DVM?
> Items marked ? are considered uncertain and a question for the group.
More information about the volt-nuts