[volt-nuts] 3478A - Kelvin Klips
brooke at pacific.net
Mon Jan 3 22:09:24 UTC 2011
A lot depends on how low. I did a lot of testing using the 34401A DMM
in 4-wire mode and determined that the DC method is not accurate below
50 milliohms because of voltage generation. This testing was done using
the HP 11059A Kelvin lead set.
Links to Kelvin clips and part numbers at the bottom of the above page.
For 100 milliohms and lower you need to use an AC method, like the 4328A
J. Forster wrote:
> A bunch of people on eBay sell knock-off Kelvin Klips. I'd like to get a
> set for 4-terminal Ohms with my 3478A.
> Does anybody have any experience on which are good and which are trash? I
> do have a slight preference for made-up lead sets w/ Bananna Plugs on the
> meter end.
> I am not interested in 8 digit accuracy, just being able to measure low
> ohms w/o contact issues.
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/volt-nuts
> and follow the instructions there.
More information about the volt-nuts