[volt-nuts] Solartron 7081 Calibration Update
J. L. Trantham
jltran at att.net
Sun Nov 13 23:38:47 UTC 2011
Forgot to mention. My workshop is a room I added off the garage when we did
a house renovation about 10 years ago. It has Heat and A/C, wired with lots
of outlets and sits about 25 to 26 degrees C year round. Right now, it's
25.0 degrees C. If I moved my shop temp to 22 degrees C, I think I would be
The calibration was at 22 degrees C, 50% humidity. I'll have to get a
'barometer' for the shop to add the rest of the data points.
From: volt-nuts-bounces at febo.com [mailto:volt-nuts-bounces at febo.com] On
Behalf Of gsteinba52 at aol.com
Sent: Sunday, November 13, 2011 10:56 AM
To: volt-nuts at febo.com
Subject: Re: [volt-nuts] Solartron 7081 Calibration Update
Please keep us in the loop for this continuing story. Free (!) Repair/Cal,
eh - maybe I should send my unit in before they figure things out...
What sort of environment do you have. Most gear, and particularly one at
this performance level, is cal'ed at a specific ambient
temperature/humidity. If you're using it as a transfer standard you should
take this into account.
More information about the volt-nuts