[volt-nuts] Trimming the LTZ1000 tempco
Andreas_-_Jahn at t-online.de
Fri Oct 7 18:00:20 UTC 2011
> However the tempco matching isnt perfect with a "typical" (note sample
> size = 1!!!!) residual of +50ppm/K (+350uV/K)
I would not talk of a sample size = 1.
My LTZ #1 has 48ppm/K with 70k and 52ppm/K with 50K
My LTZ #2 has 54ppm/K with 70k
Both have a setpoint divider of 12K5/1K.
from the datasheet you can calculate for the 13K/1K nominal temperature
100 ppm in voltage divider change R4/R5 will give 1 ppm output change.
100 ppm change of 0,5V VBe will give around 50uV temperature setpoint
2mV VBE/K or 50uV/2000uV/K = 0,025K temperature setpoint change.
So from datasheet with nominal divider you can calculate
40 ppm / K typical temperature gradient.
>From Mickles "LTZ-guru" we can calculate 39ppm/K for the
unheated reference. (0.95ppm output for 100ppm divider change)
So there seems to be less tempco (but faster ageing)
with higher temperature setpoints.
With best regards
More information about the volt-nuts