[volt-nuts] Fwd: Solartron 7150plus - Chasing AC gremlins
David C. Partridge
david.partridge at perdrix.co.uk
Fri Dec 7 09:25:22 UTC 2012
I don't think that is likely, as Phil clearly says that all the DC ranges are correct, it is just the AC ranges that are a problem. If the EAROMs were bad, you'd lose all the calibration constants.
From: volt-nuts-bounces at febo.com [mailto:volt-nuts-bounces at febo.com] On Behalf Of Wim de Jong
Sent: 07 December 2012 08:04
To: volt-nuts at febo.com
Subject: [volt-nuts] Fwd: Solartron 7150plus - Chasing AC gremlins
I suspect it to be the EAROM(s) IC 105/6 of this meter. Losing its calibration constants.
Wim de Jong
wim.de.jong.59 at solcon.nl
More information about the volt-nuts