[volt-nuts] 7081 resistance accuracy issue
m1k3k1 at hotmail.com
Sun Jan 20 04:05:57 EST 2013
As part of my checking over of the behaviour of the 7081 I ran "mode,testk" and "nines=8" overnight, the result is a swing larger than I would expect!
this means a 25 ppm shift
The circuit is meant to generate a constant current, so if that constant current can vary by 25ppm over a 20-23 degree house temperature then it is only accurate to 25ppm for resistance measurement except when used as a transfer standard.
anyone else find a similar shift?
More information about the volt-nuts