[volt-nuts] Fluke 731B battery & charge circuit
orin.eman at gmail.com
Mon Jun 17 02:48:39 EDT 2013
On Sun, Jun 16, 2013 at 11:20 PM, Christopher Brown <cbrown at woods.net>wrote:
> You might want to check/replace the current limiting resistor in the
> batt charge circuit, and the zener.
I'll check the current limiting resistor for sure. It's a carbon comp, so
it could well have drifted.
> Both of my 731Bs had low zener output.
> Had to replace the zeners in both and that 5ma jfet regulator in one.
The E505 has already been replaced with I presume a J505 as it's in a TO92
case. Do you remember what your good zener voltage was? I know they are
spec'd at +/- 5% and the J505 is +/- 20% (don't know about the E505, I
can't find a datasheet) so there could be quite a range. There is no
indication on the part list that they were selected. I have some zeners on
order and I'm going to see what I get with 1mA through them.
Thanks for the info.
More information about the volt-nuts