[volt-nuts] Seek full Manual - EDC DC millivolt calibrator E100C
cblack at centurytel.net
Sun Mar 17 22:29:37 EDT 2013
Too bad your so far away since I wouldn't mind comparing your 10v
standard against my Datron 4910 just for the fun of it.. The 4910 was
calibrated by Fluke about a year ago. All four channels ha.ve been
stable within a few microvolts in my lab over the last year. So far it
hasn't been compared to the outside world though and I really should do
some comparisons to be sure that everything is OK.
On 3/17/2013 12:32 PM, cfo wrote:
> I just bought a EDC DC millivolt calibrator E100C ,
> is $215 a fair price ?
> I expect to get it this week.
> I think it's the same as the Kron-Hite E100RC.
> The KronHite-Manual says it stops after section3
> (and $$ for the full manual) , the other has to section 4.
> See http://tinyurl.com/capynjd
> So i was wondering anyone has the full manual for either the KH E100R
> or even better the EDC E100C
> Any hints or quirks , also much appreciated.
> I have an Agilent 34401A , and a Geller 10v ref (6 month old) , so i hope
> i can verify it.
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/volt-nuts
> and follow the instructions there.
More information about the volt-nuts