[volt-nuts] FET HP3456A
tmiller11147 at verizon.net
Tue Apr 1 23:03:23 EDT 2014
Borrow Q108 (or any of the like devices) to test your hypothesis. It is just
used as a low leakage protection diode. You can replace Q108 with almost any
----- Original Message -----
From: "Tony Greene" <tonygreen924 at inbox.com>
To: <volt-nuts at febo.com>
Sent: Tuesday, April 01, 2014 10:13 PM
Subject: [volt-nuts] FET HP3456A
>I have been troubleshooting a HP3456A voltmeter.
> Its giving a stange fault, but I beleive I have nailed it down. The unit
> is reading +10.00000 volts input at +9.99661 volts. If you reverse the
> input, it reads correct at -10.00000 volts. All other ranges read correct
> on either polarity.
> What I need is a source for the FET A20 Q109 or a suitable substitute.
> Its HP part number 1855-0298, it was made originally by USA mfg 02883 -
> Siliconix, Inc, Santa Clara, CA. It crosses to NSN 5961-01-135-1324. And
> Vishay has bought them and the part number is now obselete.
> FREE 3D EARTH SCREENSAVER - Watch the Earth right on your desktop!
> Check it out at http://www.inbox.com/earth
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the volt-nuts