[volt-nuts] 34401A Why 10M ohm default i/p resistance?
Andreas_-_Jahn at t-online.de
Thu Apr 10 15:27:13 EDT 2014
perhaps its just to save the lifetime of the input range selection
relays to at least the warranty time.
Just a guess.
With best regards
Am 10.04.2014 19:58, schrieb Joel Setton:
> I think the 10 Meg default value became a de facto standard at the
> time of VTVMs (vacuum-tube volt meters), as a convenient value which
> reduced input circuit loading while remaining compatible with the grid
> current of the input triode. Designers of early solid-state voltmeters
> merely decided not to change a good thing.
> Just my $0.02 worth!
> Joel Setton
> On 10/04/2014 18:55, Steven J Banaska wrote:
>> As Tom said the 10M input impedance is used for the high voltage ranges
>> because it is a resistive divider (9.9M/100k) that can handle high
>> without much drift. Caddock THV or HVD are fairly common in precision
>> Typically you will find a high impedance (10G) path that can be used for
>> the ranges 10V and lower, but the 10M divider can be left connected and
>> will work for any voltage range by changing which side you measure.
>> As you
>> mentioned there can be an accuracy sacrifice when you have a high output
>> impedance from your source. I'm not sure why 10M is the default other
>> it may extend the life of the relay that switches the 10M divider in
>> or out.
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the volt-nuts