[volt-nuts] LM399 Long term drift specification
m1k3k1 at hotmail.com
Mon Sep 15 13:35:22 EDT 2014
On 15/09/2014 17:57, Tony wrote:
> On 14/09/2014 18:29, Andreas Jahn wrote:
>> By the way:
>> By accident I have found some ageing drift data of REF5050 on
>> Malone's homepage:
>> For me it looks around 120 ppm drift within the first year.
>> Unfortunately there is no humidity data with the plot.
>> With best regards
> That's a very interesting find Andreas, thanks. After approx 500
> hours, maximum drift is approx 80ppm, but its impossible to deduce how
> much of that is due to long term drift, temperature and humidity changes.
> The long term drift specification for the REF5050A tested is 125ppm
> (0-1000 hours), 45ppm (1000 to 2000 hours). Those for the automotive
> qualified REF5050-Q1 part are 100ppm and 10ppm respectively - way
> better after the initial 1k hours.
> Tony H
One thing that concerns me is that all 5 units seem to drift in sync, so
how much of that is the DVM doing the measuring? or is it just a single
batch with the same date code?
More information about the volt-nuts