[volt-nuts] Calibration of 34401A
illya at xdevs.com
Wed Oct 12 07:32:34 EDT 2016
I had fixed 34401A before and DIY calibrate using on-hand sources, with
calibrated 7.5d DMM as reference.
Whole process can be visible on video -
https://www.youtube.com/watch?v=9kukXTqhzsU if you like.
So it's not a rocket science, but does need set of stable
12/10/16 17:53, Dr. David Kirkby (Kirkby Microwave Ltd) пишет:
> On 12 Oct 2016 10:31, "Poul-Henning Kamp" <phk at phk.freebsd.dk> wrote:
>> In message <120112818.2234439.1476263954443 at mail.yahoo.com>, Bruce
> Griffiths wr
>>> Does anybody know if a 34401A can be adjusted by a competent 3rd party
>>> or are the details of how to adjust the meter not available outside of
>> The service manual has the calibration procedure, and it is all software.
>> Poul-Henning Kamp |
> Cheers. It seems that the lab is incompetent then, as none of the values
> appeared to be in a massive difference
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/volt-nuts
> and follow the instructions there.
More information about the volt-nuts