[volt-nuts] Fluke 332B
wpgold3637 at att.net
Sat Jul 8 10:03:49 EDT 2017
The 332D and the 332B are identical except for the accuracy specs for
the stability of the output voltages.
332B spec for 10 Volts is +/- .002% (20 ppm) for 90 days while the 332D
spec for 10 Volts is +/- .001% (10 ppm) for 60 days. The monthly drift
specs are also slightly better on the D than the B.
I suspect that FLUKE simply aged and selected a more stable reference
device along with matching the resistors in the sampling string for less
drift. For all of that you paid US$2,595 for the 332B and US$2,995 for the
332D in 1973.
I seem to remember someone replacing the chopper assembly with something
a little more modern. Don't know where I saw that. I have a B and a D and
both seemed to be very stable with the existing chopper assembly.
One thing I did find is that the electrolitic caps have a high tendency
to go bad. Not the big ones on the raw supply but those on the various plug
in assemblies. I seem to remember that they were in a range of 1 ufd to 47
ufd or so. So check them all or better yet just recap. You will save
yourself a LOT of troubleshooting time in the long run. There is also a
discussion of this subject either on VoltNuts or EEVBLOG.
----- Original Message -----
From: "David C. Partridge" <david.partridge at perdrix.co.uk>
To: "'Discussion of precise voltage measurement'" <volt-nuts at febo.com>
Sent: Saturday, July 08, 2017 5:09 AM
Subject: [volt-nuts] Fluke 332B
> Amongst the stuff I recently acquired is a Fluke 332B (sn 4151).
> I found the manual for the 332D on the Fluke website, but can't find one
> specifically for the 332B apart from an incomplete US Army version.
> I have some questions:
> 1) Does anyone know where I can find the matching manual
> 2) If not, how useful is the 332D manual?
> 3) It was last calibrated in 2010, and appears to be consistently low in
> output (about 5mV at 100V, 25mV at 500V, 50mV at 1000V) after being on for
> an hour. Is that the sort of error I should expect after that period?
> 4) I'm leery of the chopper amp, did anyone (Dr. Frank?) do a retrofit
> a modern chopper op-amp or similar?
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the volt-nuts