[volt-nuts] Best way to measure micro Ohms
csteinmetz at yandex.com
Tue Sep 19 15:32:20 EDT 2017
> Thanks. You have confirmed what I was thinking - it is *probably* the
> oxide causing the problem.
> It's not a waveguide in the normal sense of the word, transmitting a TE or
> TM wave down a hollow tube, but more like a coaxial line transmitting
> something close(ish) to a TEM wave. The outer conductor is uncoated
> aluminum and rectangular in cross section. The inner conductor is brass.
> See pictures attached
Looking at the pix, there appear to be lots of aluminum joints due to
the "built-up" construction, maximizing the potential for the sort of
troubles you are having. I would re-make the piece in brass, doing
everything possible to use as few pieces as possible (for example, by
milling recesses into one piece rather than building up a compound piece
from more easily machined sub-parts).
I would also plate everything in silver.
More information about the volt-nuts