[time-nuts] Where does the Z3801A 1 PPS come from?
Tom Van Baak
tvb at leapsecond.com
Mon Mar 14 11:51:34 EST 2005
Are you logging your lab temperature and mains
voltage? Your plot continues to intrigue me.
So I started a 53132A TI test between the 1 PPS
and the 10 MHz outputs of my main HP 58503B
GPS SmartClock (which is similar enough to a
HP Z3801A for a first test, at least).
While your plot shows about 1 ns ramp over half
a day mine so far is flat; the 5 minute averages
have a standard deviation of 15 ps. Plot attached.
I'll switch to using a Z3801A and 5370B like you if
necessary but again 1 ns over a day smells like a
tempco or voltco - in either the Z3801A or 5370B.
-------------- next part --------------
A non-text attachment was scrubbed...
Size: 15750 bytes
Desc: not available
Url : http://www.febo.com/pipermail/time-nuts/attachments/20050314/3aa0d844/attachment-0004.tiff
More information about the time-nuts