[time-nuts] Basic TIC measurement
jgray at zianet.com
Wed Oct 7 17:23:09 UTC 2009
Actually, when I got the 8140 a few years ago, I modded it to remove
the DC from the rear output. It was designed to power external
modules, which I don't have and don't need.
Somewhere I have a copy of the manual, but I downloaded yours in case
I can't find mine.
On Wed, Oct 7, 2009 at 2:37 AM, <GandalfG8 at aol.com> wrote:
> Morning Joe
> Don't forget either that the 8140 as supplied has a DC supply for the line
> taps superimposed on the rear panel sinusoidal outputs.
> However, the A4 distribution amp/output PCB was also used in several other
> Spectracom products, including the 8130 and 8164, with component
> identification always remaining the same, as far as I can tell anyway, but with link
> placement and filter values changing to suit the application.
> If you compare, for example, the A4 schematics from the 8140 and 8164
> manuals you will see that removing the DC bias is quite straightfoward.
> It would also be straightforward to set the divider output links on the
> 8140 A4 as per the 8164 A4 to provide the four different sinusoidal output
> frequencies to the rear panel although three of the four output filters would
> also need to be modified to suit the changed frequencies.
> The good news is that all the required component values are also in the
> 8164 manual:-)
> I assume you have the 8140 manual but if not it's available for download
> from the Spectracom site, a simple registration is required, or I've uploaded
> both for anyone that might be interested.....
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts