jgray at zianet.com
Sat Oct 10 06:46:07 UTC 2009
I have a 100MHz scope, DVM, frequency counter, signal generator, etc.
Sorry, no spec an. I wish I had one.
The first problem is that it fails the first performance test. Instead
of 100nS, I get about 10nS and that slowly increases to about 12nS.
This is with the unit already warmed up and it is repeatable. I
checked the internal reference output and it is very close to 10MHz,
so that isn't the problem.
Before I go any further, I think I'll open it up reseat all the
boards. It couldn't hurt.
On Fri, Oct 9, 2009 at 11:51 PM, Bruce Griffiths
<bruce.griffiths at xtra.co.nz> wrote:
> Joseph Gray wrote:
>> I just got a 5370A locally and am going through the performance
>> checks. Things are not as they should be. Is there someone who knows
>> this unit intimately who would be willing to help me off list?
>> Joe Gray
> What do you have in the way of test gear?
> A 100+MHz scope would be useful as well as a spectrum analyser.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts