[time-nuts] 3048A software question
saidjack at aol.com
Fri Feb 17 17:10:20 UTC 2012
Got the spec lines working now, thanks!
Yes, the DOS version. I was hoping for an updated user manual for that version as the manual describes the other software version, and maybe a slightly later sw version. Oh well.
Thanks for you help!
On Feb 17, 2012, at 5:06, Adrian <rfnuts at arcor.de> wrote:
> are you using the Opt. 311 (MS-DOS) software?
> I don't know of any more recent versions.
> To answer your initial question:
> You can not change anything on the diagram _after_ the measurement, not even the title.
> The spec lines will be visible only when you have defined them prior to starting the measurement.
> SAIDJACK at aol.com schrieb:
>> Hello team,
>> I am trying to get an HP 3048A system up and running, and I am having
>> problems with enabling spec limit lines on the graph.
>> I can enter the spec line data under the "manipulate results" then "spec
>> lines" menu, but the lines I entered are not visible. There are the standard
>> spec lines visible when I do the internal noise floor test, so I know this
>> should work.
>> Does anyone know how to enable these properly? The user manual doesn't talk
>> about this, and it is written for an older version of the software
>> Also, I have software version A 01.01 from 1994, does anyone have any more
>> recent software version?
>> Thanks in advance,
>> time-nuts mailing list -- time-nuts at febo.com
>> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
>> and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts