[time-nuts] Sine to square wave converter Phase Noise
David C. Partridge
david.partridge at perdrix.co.uk
Tue Jul 29 07:38:06 EDT 2014
Further to my previous post:
I note that the LTC6957 is 3.3V CMOS rather than 5V. Do you have a
recommmendation for a level converter to 5V logic? Would a 74AC04 using a
5V supply cope well enough in that role or is there a better solution?
From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com] On
Behalf Of bruce at ko4bb.com
Sent: 29 July 2014 12:00
To: Discussion of precise time and frequency measurement
Subject: Re: [time-nuts] Sine to square wave converter Phase Noise
Sorry, I meant the LTC6957-4 of course._SELA = l FILTA = L FILTB = H ie an
input stage BW of 160MHz as recommended for a 10MHz + 10dBm input.
Input for the HX4210 was +14dBm.
Comparator was that used in David Partridge's programmable divider (includes
> On July 29, 2014 at 5:37 AM Magnus Danielson
> <magnus at rubidium.dyndns.org>
> You alternate LTC7957-4 and LTC6957-4 and it is only the later that
> The LTC6957-4 has a single-stage amplifier stage input, with somewhat
> programmable bandwidth. I assume you used the 50 MHz BW setting.
> On 07/29/2014 10:55 AM, bruce at ko4bb.com wrote:
> > Recently I have been comparing the phase noise of the HOlzworth
> > HX4210 and the LTC7957-4.
> > I have found that the performance of these devices is comparable
> > with a 10MHz sinewave input with a PN noise floor below -160dBc/Hz.
> > For offsets below 100Hz the LTC6957-4 is quieter and for offsets
> > below 50Hz or so is lower than that of my measurement setup.
> > I merely mounted the LTC evaluation board in a diecast metal box to
> > reduce the effect of air currents on the LTC7957-4 and used an ultra
> > low noise 3.3V power supply (Abracon) to minimise the power supply
> > contribution to the measured phase noise.
> > Achieving a phase noise performance equal to that on the datasheet
> > was relatively easy.
> > In fact for low offsets the phase noise is lower than that shown on
> > the datasheet.
> > I've also measured the phase noise of a comparator based sine to
> > CMOS converter and its PN floor is about 20dB higher than that of
> > the LTC6957.
> > On completion of measurements PN plots for the various sine to
> > square conveters tested will be added to the web pages I maintain on
> > Didiers site..
> > Bruce
> > _______________________________________________
> > time-nuts mailing list -- time-nuts at febo.com To unsubscribe, go to
> > https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> > and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com To unsubscribe, go to
> and follow the instructions there.
time-nuts mailing list -- time-nuts at febo.com To unsubscribe, go to
and follow the instructions there.
More information about the time-nuts