[time-nuts] 3.0GHz Channel 3 installation in Agilent 53132A counter
actast at hotmail.com
Tue Mar 18 15:39:55 EDT 2014
I think the solution will be in the firmware. It must be factory set so default is the 5GHz C channel option. The 12GHz option is the same part as the 5GHz sampler with a different firmware lock.
> From: jsrobbins at earthlink.net
> To: time-nuts at febo.com
> Date: Tue, 18 Mar 2014 14:19:14 -0400
> Subject: [time-nuts] 3.0GHz Channel 3 installation in Agilent 53132A counter
> I hope this is not too OT.
> I have acquired a 3.0GHz original Channel 3 board for my Agilent 53132A
> counter. When installed, it reads four times (4x) the actual frequency.
> I'm wondering if anyone has any experience with such an installation and can
> point me to some setup item or zero ohm resistor I'm supposed to install.
> All that the manual says is to have it done by the factory. That will cost
> me more than the board itself. Many thanks.
> Jim Robbins
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts