[time-nuts] need advice on
magnus at rubidium.dyndns.org
Sat Mar 22 10:15:18 EDT 2014
On 22/03/14 12:48, Claude Fender wrote:
> I have two generators 3324A and 33120A and one counter 5334B. All the instruments are locked to a GPSDO.
> I've measures a 1 kHz output of the generators with the counter set with a gate time of 1 second, and repeated the measure 3600 times.
> The average for the 3324A is 999.9999996 Hz with a standard deviation of 6.79E-5
> The average for the 33120A is 999.9999986 Hz with a standard deviation of 1.04E-5. (see joined graph)
> The resolution of the 3324A is 1 mHz, the resolution of the 33120A is 0.1 mHz
> So here is my question ! Do you think if I find a generator with a resolution less than 0.1 mHz, the repetability will be better ? If so, do you have any advice for some stuff less than 1000 USD ?
> Thanks in advance !
The 3325A/B has 1 uHz resolution.
Be careful when making frequency measurements, as some counters can
create a small measurement bias. Using TI-mode and TimeLab you can avoid
More information about the time-nuts