[time-nuts] Isotemp OCXO107-10 Internal Photos
ed_palmer at sasktel.net
Tue Mar 25 23:54:10 EDT 2014
FYI, I've posted a few pictures of the inside of this oscillator.
Noteworthy is the tiny Dewar flask.
If you click on the magnifying glass at the bottom of the picture and
then do it again, you'll get the full size picture.
After about a week of operation, aging was < 5e-10/day (spec is <
2e-10/day). An impressive result for such a short run time. Adev with
drift removed at Tau > 100 sec. is stuck at ~1e-11 as measured against a
Tbolt. I can't find a spec for it. The Tbolt and measurement system
are capable of much better performance than that so the Isotemp appears
to be the limiting factor. Maybe longer run time would bring it down.
More information about the time-nuts