[time-nuts] 53230A noise floor
anders.e.e.wallin at gmail.com
Fri Apr 3 05:49:15 EDT 2015
I ran some tests on a new 53230A counter that just arrived:
The time-interval single-shot noise and resulting ADEV seems easy enough to
I didn't yet have time to look closely at Enrico Rubiola's notes over here:
There is probably a good explanation for the ADEV-level in standard
(pi-counting?) reciprocal frequency counter mode as well as the roughly
1/sqrt(10) enhancement in ADEV when increasing the gate-time 10-fold.
on another note:
The discussion on phase-noise <-> ADEV calculations earlier was
interesting. Datasets (synthetic or real) with known correct ADEVs and
phase-noise-spectra would be a good addition to allantools and a motivation
to add code for the phase-noise <-> ADEV conversion.
More information about the time-nuts