[time-nuts] Has my TBolt gone south ?
bpl521 at outlook.com
Wed Aug 12 21:50:46 EDT 2015
I have had a similar issue occur. For some reason it all of a sudden wanted to do wanted to do a self survey but couldn't. Stuck at 1% on the survey. I switched to Tboltmonitor as my backup program and was able to re input my lat/long and altitude settings, saved and it carried on. Power glitch, firmware issue, who knows.
p.s. I believe you need a minimum of three satellites to work through the standard or precision survey.
> To: time-nuts at febo.com
> From: hmurray at megapathdsl.net
> Date: Wed, 12 Aug 2015 11:29:58 -0700
> CC: hmurray at megapathdsl.net
> Subject: Re: [time-nuts] Has my TBolt gone south ?
> dibene at usa.net said:
> > Today I switched on again my TBolt after a few weeks of power down. It looks
> > like it has decided to take a period of rest, maybe definitive...
> How long did you wait after turning it on?
> How good is your antenna position?
> Did you tell it to remember the location after a survey finished?
> If it's been off for a few weeks, it doesn't know where the satellites are.
> To get started, it has to find one by a brute force search. With a good
> antenna, that doesn't take very long. Then it takes 15 minutes to collect
> the satellite info. With a poor antenna location, it takes longer, maybe a
> lot longer.
> These are my opinions. I hate spam.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts