[time-nuts] HP5065A C-field current is temperature sensitive
rfnuts at arcor.de
Sat Aug 29 09:03:36 EDT 2015
Keep in mind that potentiometer R6 is one of the key components of the
Together with R8 and R12, it defines the reference voltage to Q6A/B.
The tempco of that wire-wound potentiometer can certainly not be neglected.
I would suggest to check the C-field temperature dependency with an
external precision current source feeding the C-coil directly.
Poul-Henning Kamp schrieb:
> Using the Fluke 732A as voltage reference improved HP5065A stability,
> but not very much, so I looked closer at the constant-current
> generator and the C-field pots "calibration":
> ... and found out that the current-sensing resistors R10||R11 have
> a very high temp-co.
> Getting closer...
More information about the time-nuts