[time-nuts] HP5065A C-field current is temperature sensitive
phk at phk.freebsd.dk
Sat Aug 29 11:09:03 EDT 2015
In message <55E1ADA8.2000102 at arcor.de>, Adrian writes:
>Keep in mind that potentiometer R6 is one of the key components of the
>Together with R8 and R12, it defines the reference voltage to Q6A/B.
>The tempco of that wire-wound potentiometer can certainly not be neglected.
Absolutely not, it's clearly part of the picture, but something has
10x the tempco of that one right now.
>I would suggest to check the C-field temperature dependency with an
>external precision current source feeding the C-coil directly.
That's what I'm doing right now: C-field driven by Fluke 732A
through 2.5k resistor.
Poul-Henning Kamp | UNIX since Zilog Zeus 3.20
phk at FreeBSD.ORG | TCP/IP since RFC 956
FreeBSD committer | BSD since 4.3-tahoe
Never attribute to malice what can adequately be explained by incompetence.
More information about the time-nuts