[time-nuts] Has anybody checked this? GPSDO in kit
bpl521 at outlook.com
Sun Oct 30 16:44:32 EDT 2016
Yes sorry, you are right it is at RS232 levels.
From: time-nuts <time-nuts-bounces at febo.com> on behalf of Mark Sims <holrum at hotmail.com>
Sent: October 30, 2016 10:48 AM
To: time-nuts at febo.com
Subject: [time-nuts] Has anybody checked this? GPSDO in kit
No, the 4 pin connector is at RS-232 levels. It connects to an RS-232 level converter chip. These "UCCM" devices speak SCPI at 57600,8,N,1 I have Lady Heather talking to them (soon to be released, I promise). I have tested it with four different firmware versions (all have some slight differences).
Most of the NMEA data sent from the receiver is proprietary Furuno commands. KO4BB.COM has a manual for the receiver...
Looking at the fifth picture on the ebay link, at the top left corner next to the SMA connector just below R13 is a vertical row of holes. That is the main serial connection and it is at TTL levels.
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
time-nuts Info Page - American Febo Enterprises<https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts>
time-nuts is a low volume, high SNR list for the discussion of precise time and frequency measurement and related topics. To see the collection of prior postings to ...
and follow the instructions there.
More information about the time-nuts