[time-nuts] Deaf Z3801
michael.cook at sfr.fr
Wed Nov 15 03:26:25 EST 2017
> Le 15 nov. 2017 à 04:43, Bob Bownes <bownes at gmail.com> a écrit :
> I put my 3801 back in service the day before yesterday, or tried to. It’s been on the shelf for about a year (off).
> I powered it up, and while it sees a few sats, it will not acquire any. Changing to a known good antenna (off the NTP server) makes no difference.
> Anyone seen the behavior before? The location data is good, but the time/date are way out.
Probably GPS week rollover. I think the fix is to power it up with the antenna disconnected, set te date and time manually and then reconnect the antenna.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
"The power of accurate observation is commonly called cynicism by those who have not got it. »
George Bernard Shaw
More information about the time-nuts