[time-nuts] Deaf Z3801
jmfranke at cox.net
Wed Nov 15 08:49:51 EST 2017
Shut off power to the GPS receiver
Disconnect the GPS antenna downlead
Reapply power to the GPS receiver
Enter :GPS:INIT:DATE 2017,01,01 (The only space is between DATE and 2017)
Reconnect the GPS antenna downlead and wait
> On November 14, 2017 at 10:43 PM Bob Bownes <bownes at gmail.com> wrote:
> I put my 3801 back in service the day before yesterday, or tried to. It’s been on the shelf for about a year (off).
> I powered it up, and while it sees a few sats, it will not acquire any. Changing to a known good antenna (off the NTP server) makes no difference.
> Anyone seen the behavior before? The location data is good, but the time/date are way out.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts