[time-nuts] Allan variance by sine-wave fitting
Ralph Devoe
rgdevoe at gmail.com
Wed Nov 22 10:58:19 EST 2017
Hi time nuts,
I've been working on a simple, low-cost, direct-digital method for
measuring the Allan variance of frequency standards. It's based on a
Digilent oscilloscope (Analog Discovery, <$300) and uses a short Python
routine to get a resolution of 3 x 10(-13) in one second. This corresponds
to a noise level of 300 fs, one or two orders of magnitude better than a
typical counter. The details are in a paper submitted to the Review of
Scientific Instruments and posted at arXiv:1711.07917 .
The method uses least-squares fitting of a sine wave to determine the
relative phase of the signal and reference. There is no zero-crossing
detector. It only works for sine waves and doesn't compute the phase noise
spectral density. I've enclosed a screen-shot of the Python output,
recording the frequency difference of two FTS-1050a standards at 1 second
intervals. The second column gives the difference in milliHertz and one can
see that all the measurements are within about +/- 20 microHertz, or 2 x
10(-12) of each other, with a sigma much less than this.
It would interesting to compare this approach to other direct-digital
devices.
Ralph DeVoe
KM6IYN
-------------- next part --------------
A non-text attachment was scrubbed...
Name: Capture.JPG
Type: image/jpeg
Size: 125115 bytes
Desc: not available
URL: <http://www.febo.com/pipermail/time-nuts/attachments/20171122/7c42c976/attachment-0001.jpe>
More information about the time-nuts
mailing list