[time-nuts] TruePosition GPSDO Holdover Issues
kb8tq at n1k.org
Sun May 13 14:05:50 EDT 2018
When you are looking at timing, the SBAS / WAAS sat’s really don’t count as part of the total
of 4 that you need for a basic fix. There also iis the subtle distinction of “tracking” vs “locked to”
on some devices. Tracking means we might get adequate data soon and locked means it is
good enough to use on those devices.In that case, only the “locked” sats count towards the
minimum of 4 that you must have.
Past the minimum of 4 rule, most GPSDO’s also want to see that set of devices for some period
of time before they come out of holdover. You will drop in very quickly ( a second or two), but come
out slowly ( many minutes). Local noise can in some cases be enough to put you in holdover.
> On May 13, 2018, at 1:13 PM, gandalfg8--- via time-nuts <time-nuts at febo.com> wrote:
> The location at 55N, 5W, isn't ideal, there's quite a large hole to the north but this isn't something I've seen here before with any other GPS module or GPSDO.
> When first noticed I'm sure it was whilst tracking six or seven sats, it was certainly five or more, which is why I commented in the first place, it was only later I thought there might be some correlation with it tracking low numbers.
> There doesn't seem to be any adjustment for elevation mask on these, at least not via Lady H, but I've switched now from a flat survey antenna to a Symmetricom pod on a stub mast, so I can cheat a bit and angle it south slightly:-)
> It'll take some time to build up a picture of the effect of that but it's tracking 8 sats at the moment.
> Inmarsat-3, PRN120, seems to have joined in the mix now and keeps popping on and off the bottom of the list but I'm not sure whether or not that could contribute anything useful anyway.
> Nigel, GM8PZR
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts