[volt-nuts] Keithley 2001 Multimeter Fault
swbenn at gmail.com
Fri Mar 5 23:37:00 UTC 2010
I have had luck a couple of times sweeping an IR thermometer across a PCB
when the finger was not sensitive enough to find the hot part. A shorted
part that has plenty of copper soldered to it can sometimes be only a little
warmer the the nearby components.
All I need to do now is hot melt the IR thermometer to the old plotter......
Date: Fri, 5 Mar 2010 18:15:13 -0000
From: "Alan Scrimgeour" <scrimgap at blueyonder.co.uk>
Subject: Re: [volt-nuts] Keithley 2001 Multimeter Fault
To: "Discussion of precise voltage measurement" <volt-nuts at febo.com>
C768B6E7B1884FE28E1CDC415DEAE6EF at AlanPC>
Content-Type: text/plain; format=flowed; charset="iso-8859-1";
Thanks John, I plan to start the search in a few hours. Wish I had a thermal
More information about the volt-nuts