[volt-nuts] Keithley 2001 Multimeter Fault
scrimgap at blueyonder.co.uk
Sat Mar 6 01:05:41 UTC 2010
Nice idea - that might just be the excuse I need to get one!
----- Original Message -----
From: "Steve Bennett" <swbenn at gmail.com>
To: <volt-nuts at febo.com>
Sent: Friday, March 05, 2010 11:37 PM
Subject: [volt-nuts] Keithley 2001 Multimeter Fault
> I have had luck a couple of times sweeping an IR thermometer across a PCB
> when the finger was not sensitive enough to find the hot part. A shorted
> part that has plenty of copper soldered to it can sometimes be only a
> warmer the the nearby components.
> All I need to do now is hot melt the IR thermometer to the old
> Message: 3
> Date: Fri, 5 Mar 2010 18:15:13 -0000
> From: "Alan Scrimgeour" <scrimgap at blueyonder.co.uk>
> Subject: Re: [volt-nuts] Keithley 2001 Multimeter Fault
> To: "Discussion of precise voltage measurement" <volt-nuts at febo.com>
> Message-ID: <
> C768B6E7B1884FE28E1CDC415DEAE6EF at AlanPC>
> Content-Type: text/plain; format=flowed; charset="iso-8859-1";
> Thanks John, I plan to start the search in a few hours. Wish I had a
> imaging camera!
> volt-nuts mailing list -- volt-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the volt-nuts