[volt-nuts] Calibrating an HP 34702A?
Dr. Frank Stellmach
drfrank.stellmach at arcor.de
Thu Feb 24 21:52:45 UTC 2011
The basic precision of your standards is not really sufficient, as they
should have at leat 10ppm / 50ppm accuracy.
The more severe problem is, that you are lacking any transfer standards,
i.e. transfer from your 2,5V and 596 Ohm standard values to the
necessary Cardinal Points, 0,1V, 1V, 10V, 100V, 1kV, and Ohm ranges
Therefore, you have no chance to calibrate your DMM properly without
Regards - Frank
More information about the volt-nuts