[volt-nuts] 7081 connector part number Q
m1k3k1 at hotmail.com
Wed Apr 18 14:47:39 UTC 2012
> Date: Wed, 18 Apr 2012 16:16:21 +0200
> From: fabioeb at quipo.it
> To: volt-nuts at febo.com
> Subject: Re: [volt-nuts] 7081 connector part number Q
> Hello, followed the discussion about connector, I don't want to
> spend more on the connectors than on the meters I have.
Yes, it seems painful, but the alternative is to butcher the meter.
> I'm thinking to build some, I have a pair of cheap kelvin clips,
> the ones used on RLC meters with 4bnc connectors.
> The bnc central pin fits in the solartron connector and also bnc
> outer shell fits in, so I can use one shell and 5 pins to build
> a connector, I need a way to fix in place the pins.
> Each pin will be soldered to cable and isolated by heat shrink tubing,
> after it will be inserted in the connector and I will use
> a syringe to inject some glue to fix them in place, naturally
> the connector will be held vertical and the bottom of the connector
> will be sealed.
The way to do it is to machine a part like Mickle did.
> I'm thinking about some two component glue like uhu plus
> in the connector, what do you think about this, will I have leackage
I think it is quite likely, the specified parts are made with a PEEK insulator.I remember having a teflon part machined for the company I was working for and the subcontractor used an inappropriate cutting fluid so the teflon was badly conductive. a week in the ultrasonic bath with HPLC grade 99.999% IPA didnt cure them, they had to be scrapped.
> "David C. Partridge" <david.partridge at perdrix.co.uk> ha scritto:
> > Be prepared for pain. When I last got some a couple of years ago
> > they were about £25 plus VAT per connector, with a MOV of about £100.
> > Dave
More information about the volt-nuts