[volt-nuts] HP3458A - Ram Test 1 Low
phk at phk.freebsd.dk
Sat Dec 15 18:17:35 UTC 2012
In message <1355594832.29796.YahooMailNeo at web164002.mail.gq1.yahoo.com>, Lawren
ce Sawicki writes:
>Awww. I am thinking that one of my NVRAMs gave up in my HP3458A.
At least it does not seem to be the CAL RAM
Try leaving the meter powered on for some time, then power cycle
If that works, make a backup copy of your CALRAM immediately
(See old message from me in mailling list archive for how to do
this across the GPIB interface)
Poul-Henning Kamp | UNIX since Zilog Zeus 3.20
phk at FreeBSD.ORG | TCP/IP since RFC 956
FreeBSD committer | BSD since 4.3-tahoe
Never attribute to malice what can adequately be explained by incompetence.
More information about the volt-nuts